The Therapeutic Goods Administration (TGA), the Australian authority responsible for medical devices regulation, issued guidance highlighting performance requirements applicable to ventilators allowed to be used in Australia during the outbreak of the Coronavirus Disease 2019 (COVID-19) caused by the virus “SARS-CoV-2” or the “novel coronavirus” resulting in increasing demand on ventilators due to the respiratory illnesses and complications caused by the virus. The scope of the document covers the ventilators to be used for adult patients except those having complex responses when special devices should be used. It is also important to mention that rules and requirements set forth by the present specification should not be applied to the devices already approved to be marketed and used in Australia. The administration also emphasizes the importance of using any complex medical equipment, such as ventilators, by experienced and properly trained healthcare professionals.
TGA Requirements on Features of Ventilators
In particular, the document from the TGA describes the minimal requirements the device should meet in order to be allowed for marketing and use in Australia. First of all, the specification outlines the features the ventilator should have to be approved for use for the intended purpose. The list of the features the document contains includes, inter alia, the following ones:
- Mandatory ventilation. The device should allow healthcare professionals to use it to set a respiratory rate, and also a tidal volume. The ventilator should also provide the possibility to switch between pressure-controlled ventilation (PCV) and volume-controlled ventilation (VCV).
- Positive and expiratory pressure (PEEP) should be adjustable in steps of 2cm H2O or smaller, providing the general range of 5 – 25 cm H2O.
- Respiratory rate should be adjustable is steps of 2 or less with the general range 5 – 30 breaths per minute.
- Tidal volume should be adjustable is steps of 50 ml or less, with the time-varying from 0.5 to 2 seconds.
- Displays should provide information on settings of inspiratory time, frequency, Vt (tidal volume), PEEP and ventilation mode. It is also allowed to use mechanical displays on control devices.
- Alarms. The ventilator should generate alarms in case of failure of electricity or gas supply, switching of the machine when connected to a patient, and in other cases prescribed by specification.
Besides technical and operational requirements described from the TGA hereabove, the device should also meet the following additional requirements:
- The labeling of the devices should contain information about all functions and controls provided in generally accepted terms and pictograms,
- The device should be supplied with the detailed instructions for use providing healthcare professionals using it with the exhaustive information necessary to ensure the safety and performance of the device as intended by the manufacturer, and also the recommendations on fixing possible errors and malfunctions,
- The design of the device, its manufacturing process, and supply chain should be transparent enough for the regulating authority to access its reliability,
- The construction of the device should provide the possibility to move it,
- It should be possible to perform general cleaning and disinfection procedures applicable to the devices of this type,
- All disposable elements and spare parts should be available in Australia (this requirement is especially important due to the freight movement restrictions imposed nowadays).
The specification issued by the TGA also provides detailed requirements on raw materials to be used, the manufacturing process and the hazard mitigation measures to be applied by the manufacturer to reduce the identified potential risks. These requirements include the following ones:
- The materials used should be simple and pure when it is reasonably possible.
- It is recommended to use polyethylene and polypropylene as materials for flexible components.
- The use of additives should be avoided to the maximum extent possible.
- The use of polyvinyl chloride (PVC) should be avoided, especially for the patient gas pathway.
- Ventilators should be manufactured in a reasonably clean room while it is not required to comply with the cleanroom specifications.
- The design of the device should provide HME filters placed between the ventilator and breathing system to avoid contacting with the chemicals removed from the device during its use.
It is also stated that ventilators should meet the requirements set forth in the applicable guidelines issued by the Medicines and Healthcare products Regulatory Agency (MHRA), the UK regulating authority. In particular, the guidelines on testing protocols for final validation of safety and performance of RVMS should be applied, including the appropriate appendix thereto concerning the software that could be determined as a medical device.
Ventilators` Desirable Features
Besides the mandatory features and requirements described by the TGA, the document also provides a list of additional features the authority recommends to implement. The list of recommended features includes the following ones:
- Spontaneous ventilation. Depending on whether the patient starts to breathe in or breathe out, the device should automatically switch between the inspiratory pressure and expiratory pressure. At the same time, in case if the patient stops breathing in pressure support mode, the device should automatically switch to mandatory ventilation mode.
- Synchronized mandatory ventilation – special mode providing that the device synchronizes the mandatory ventilation to the efforts made by the patient. The device should also provide support for patients attempting to breathe themselves.
- Exhalation filters. It is advised to use removable elements to simplify cleaning and disinfection procedures.
- Batteries should allow the device to be used for 2 hours.
- Displays. The device should display the confirmation of each patient breath when operating in pressure support mode, and also notify healthcare professionals when it becomes below the acceptable range.
The document from the TGA also outlines the applicable international harmonized standards the device should meet, including the following ones:
- ISO 10651-3:1997 Lung Ventilators for Medical Use – Emergency and Transport.
- ISO 80601-2-84:2018 Medical electrical equipment.
- ISO 19223:2019 Lund ventilators and related equipment.
- AS 2896-2011 Medical gas systems – Installation and testing of non-flammable medical gas pipeline systems.
- IEC 60601-1-11:2015 Medical electrical equipment – General requirements for basic safety and essential performance.
- IEC 62353:2014 Medical electrical equipment – Recurrent test and test after repair of medical electrical equipment.
Summarizing the information provided hereabove, the specification issued by the Australian TGA describes the minimum requirements ventilators should meet in order to be allowed to be placed on the market, and also the additional features and functions the authority recommends to implement to improve the safety and performance of the devices. The document also provides a list of applicable standards medical device manufacturers could refer to with regard to the design and composition of the devices. By issuing the present specification, the authority intends to find a balance between ensuring the highest level of patients` health protection and making ventilators available in Australia in the amount sufficient to meet the increased demand these devices face nowadays due to the respiratory illnesses and complications caused by the COVID-19.
How Can RegDesk Help?
RegDesk is a next-generation web-based software for medical device and IVD companies. Our cutting-edge platform uses machine learning to provide regulatory intelligence, application preparation, submission and approvals management globally. Our clients also have access to our network of over 4000 compliance experts worldwide to obtain verification on critical questions. Applications that normally take 6 months to prepare can now be prepared within 6 days using RegDesk Dash(TM). Global expansion has never been this simple.